Views:0 Author:Site Editor Publish Time: 2021-11-11 Origin:Site
Due to the impact of the epidemic, nucleic acid extraction system becomes more popular. In order to make the installation more convenient for customers who purchase the instrument, the following will introduce the installation and debugging of the nucleic acid extraction system.
1) Place the instrument
Nucleic acid extraction system is only suitable for indoor installation with good indoor ventilation, no corrosive gas, and no strong magnetic field interference.
Do not place the instrument in a humid or dusty place;
do not place the instrument on a soft cushion, so as to prevent the base from sinking into the soft cushion and block the vent below;
the openings on the instrument are for ventilation and heat dissipation.
Leaning against walls or stacking other objects, do not block or cover these ventilation holes. When the instrument is running, the distance between the ventilation holes of the instrumeand the nearest object should not be less than 25cm, so as not to affect heat dissipation; do not use the instrument in direct sunlight. Keep away from heaters, stoves and all other heat sources.
2) Ground the instrument
To avoid electric shock, the input power cord of the instrument must be grounded reliably. General nucleic acid extraction system uses a three-pin plug, which can only be used with this type of power socket. This is a safety device.
3) Power supply
1. Power requirements
Before connecting the AC power supply, make sure that the voltage of the power supply is consistent with the voltage required by the instrument (a deviation of ±10% is allowed). And make sure that the rated load of the power socket is not less than the requirement of the instrument.
2.Pay attention to the power cord
This instrument should usually use the power cord that comes with it. If the power cord is damaged, it must be replaced and not repaired. The power cord of the same type and specification must be used instead. When the instrument is in use, do not put anything on the power cord, and do not place the power cord where people move.
3.Plug and unplug the power cord
Always hold the plug when plugging or unplugging the power cord. When inserting the plug, make sure that the plug is fully inserted into the socket. Do not pull the power cord when unplugging the plug.
Our popular product nucleic acid extraction system BNP32 is an open platform. It has performance characteristics such as sample protection, modular design, and humanized interface. If you are interested in our nucleic acid extraction system, you can contact our sales manager. We are happy to serve you.